Uses the vapor power emissions cold air / 2 times. Plug in the power cord. Shut exhaust needle valve in the Close (clockwise).
Adding about 800 ml of distilled water. Close the door cover, tight the rotate handle. Start the timer knob to 121. Sterilization is complete, please be sure to open the exhaust needle valve firstly (inverse clock direction).
Wait for the pressure gauge pointer to zero, opened the door cover. Every time sterilization, must add about 800 ml of distilled water. Use a separate socket (confirm the socket of the output voltage).
Pressure gauge pointer in the red areas, please be sure to first Open the exhaust steam of the needle valve. Plate 3pcs (Size:2115.52 cm). Steam exhaust pipe 1pc(1 meter length).The sale of this item may be subject to regulation by the U. Food and Drug Administration and state and local regulatory agencies. This item is in the category "Business & Industrial\Healthcare, Lab & Dental\Medical & Lab Equipment, Devices\Autoclaves & Sterilizers". The seller is "dentistworks" and is located in this country: US. This item can be shipped to United States.